Question: What happens if a customer operates a run when the tissue lies outside the allowable area on the plain glass slide?
Answer: If the tissue is outside of the allowable area on the tissue slide, probe transfer will not occur and no data will be generated in those areas. The allowable area denotes the region for compatible probe transfer to the Visium Slide Capture Area. To ensure optimal results place sections within the allowable area or use archived slides that already have tissue placed within this area.
Please refer to the Visium CytAssist Quick Reference Card for specific guidance on tissue section location. The compatible region will differ depending on whether frosted or unfrosted glass slides are used.
Products: Visium CytAssist for FFPE, Visium CytAssist for FF