Question: Can the Chromium Controller be moved between labs?
Answer: Yes. To move the Chromium Controller or Chromium Single Cell Controller between laboratories, the instrument can be carried by hand or moved with a cart. The instrument should be handled carefully.
If the instrument is going to be shipped or moved in a vehicle, it should be repackaged in the original shipping boxes with the foam strip inserted in the tray drawer. Instructions for repacking are attached.
After moving the instrument either by hand or by shipping, we recommend re-running the Readiness Test. (The test chip, chip holder, and gaskets are included with the initial shipping of the instrument.) Instructions for running the Readiness Test can be found at: https://support.10xgenomics.com/genome-exome/index/doc/user-guide-chromium-controller-readiness-test
Products: Chromium Controller, Chromium Single Cell Controller